No.3, Jiangjunzi Road, Bangjiangxi Village, Dalong Street, Shiqi Town, Panyu District, Guangzhou City
Continuous improvement, professional attitude, teamwork
What to do if the water flow of the rain test box does not meet the standard
The rain test box is a device for testing the product shell against rain and splash water. It is generally suitable for electronic appliances, outdoor lamps, auto parts, communication equipment, drones, smart electronics, household appliances and other products.
Recently, when some customers communicate with our after-sales service department, the editor found that a problem is often mentioned, that is, during the test of the rain test chamber, the water flow cannot reach the specified standard value. This waterproof The test is not up to the standard, which seriously affects the product development progress and test results.
So why does this happen? The water flow of the rain test chamber is not up to standard. After the analysis by the after-sales engineer, the main reasons are the following. The editor also gave a solution for your reference.
1. The test water does not meet the standard. It is recommended to use pure water for testing. Samples should be treated with dust before testing to prevent impurities from being brought into the test water source and effectively inhibit the appearance of scale from the source.
2. The nozzle of the swing pipe is blocked. It is recommended to use a needle to clear the spray hole. Yuexin's IPX3/4 rain test chambers are equipped with spray holes through needles, which can help users quickly clean up scale.
3. The nozzle of the swing pipe is severely deformed. It is recommended to replace the nozzle immediately. Our IPX3/4 rain test chambers are equipped with spare nozzles, which can bring customers a good experience.
Well, the above are some small methods summarized by the editor, I hope it can be helpful to everyone. If you want to know more about the rain test chamber, please contact our customer service.
Yuexin-Focus on the R&D and production of the rain test chamber, and look forward to every communication with you.
This article is an original article by Guangzhou Yuexin Test Equipment Co., Ltd., and may not be reproduced without permission.
How to carry out container rain test